Type of Document Master's Thesis Author Bao, Jianxiong Author's Email Address email@example.com URN etd-06302005-122426 Title Negative Temperature Programming using Microwave Gas Chromatography Degree Master of Science Department Chemistry Advisory Committee
Advisor Name Title Taylor, Larry T. Committee Chair Anderson, Mark R. Committee Member Hanson, Brian E. Committee Member Keywords
- gas chromatography
- negative temperature programming
Date of Defense 2005-06-21 Availability unrestricted AbstractThe microwave region of the electromagnetic spectrum lies between the infrared radiation and radio frequency region and corresponds to wavelengths of 1 cm to 1 m (frequencies of 30 GHz to 300 MHz respectively). Gaisford and Walters developed the microwave GC oven in 2000, which operated at 919 MHz. The microwave GC oven has been engineered to generate a uniform microwave field around an open tubular capillary column with elimination of cold spots which are common in a domestic microwave oven. Only the column is heated in a microwave GC oven. In microwave GC, ramp rates in excess of 6 degree/sec, isotherms as high as 450 degree, and 1 min cool-down times from 300 to 50 degree can be achieved. In conventional GC, only positive temperature programming is used because the oven cools down slowly due to the great thermal mass of the oven. Short cool-down time in microwave-heating GC makes it possible to employ negative temperature programming for the separation of compounds during the process.
The theory and feasibility of negative temperature programming in microwave GC were investigated in this research. Negative temperature programming was employed to analyze and quantitate four different critical pairs of non-volatile and volatile compounds. The influence of cooling rate, holding time, and reheating rate in negative temperature programs for enhanced resolution were investigated. The results obtained from negative temperature programming were compared to those from positive temperature programming. Negative temperature programming afforded greater resolution of critical pairs of analytes.
Filename Size Approximate Download Time (Hours:Minutes:Seconds)
28.8 Modem 56K Modem ISDN (64 Kb) ISDN (128 Kb) Higher-speed Access JXBAO01.pdf 35.51 Kb 00:00:09 00:00:05 00:00:04 00:00:02 < 00:00:01 JXBAO02.pdf 569.13 Kb 00:02:38 00:01:21 00:01:11 00:00:35 00:00:03
If you have questions or technical problems, please Contact DLA.