Type of Document Master's Thesis Author Herrmann, Christine Author's Email Address email@example.com URN etd-07052005-170008 Title Estimating Joint Torques on a Biodex System 3 Dynamometer Degree Master of Science Department Mechanical Engineering Advisory Committee
Advisor Name Title Madigan, Michael L. Committee Chair Granata, Kevin P. Committee Member Nussbaum, Maury A. Committee Member Keywords
Date of Defense 2005-05-11 Availability restricted AbstractThe purpose of this investigation is to propose a method for determining resultant joint moments from a Biodex System 3 isokinetic dynamometer. Previous investigations using a dynamometers determined that measured moments from a dynamometer do not equate to the moment applied by the joint. Using Biodex System 3 dimensions and equipment, the proposed method corrects for gravity, acceleration and inertia moments, and relative angular movement between the dynamometer and the joint axis of rotation for the knee and ankle. The current method includes gravitations correction using a 3rd order polyfit method to a 4°/s passive trial, and inclusion of inertial moments from the dynamometer arm and limb segment. A method is also proposed to correct for gravitational moments, acceleration and inertia moments, and distal joint moments while testing the hip.
Previously proposed methods are then compared to the proposed method in isometric and isokinetic exertions. The comparison to a known moment concluded that the results for the isometric exertion are accurate for the proposed method. If the torque measurements from the dynamometer are independent of the velocity, as reported by the manufacturer, the validation of the proposed method for isometric testing holds true for isokinetic as well. The results from isokinetic testing show reasonable results for determining the resultant joint moments.
The proposed method can be simplified for clinical or experimental testing. If inertial and acceleration moments are not of concern, than using the propsed gravitational correction will account for the COM (Center of Mass). No additional measurements of the limb segement and dynamometer attachment are needed. The proposed method is recommended for Biodex System 3 isokinetic dynamometer correction in obtaining resultant joint moments at the knee, ankle, and hip.
Filename Size Approximate Download Time (Hours:Minutes:Seconds)
28.8 Modem 56K Modem ISDN (64 Kb) ISDN (128 Kb) Higher-speed Access herrmann_irb_approval.pdf 89.90 Kb 00:00:24 00:00:12 00:00:11 00:00:05 < 00:00:01 Herrmann_Thesis_71405.pdf 3.48 Mb 00:16:06 00:08:17 00:07:14 00:03:37 00:00:18indicates that a file or directory is accessible from the Virginia Tech campus network only.
If you have questions or technical problems, please Contact DLA.